The Thermoelectric Container ТEK-3,3-12 is designed to transport drugs and organ donation.The container is made on the basis of thermoelectric modules operating on the Peltier effect.The container construction design allows it to operate from external power source and from the built-in battery. When the external power source is disconnected, the accumulator unit switches on automatically (when the line circuit breaker is switched on).
The container has a combination lock on the lid of the cooling chamber.
The set includes:
container 1 pc
power supply cord 1 pc
instruction manual 1 pc
The general view of the container is shown in Pic.1. Here: 1 – line circuit-breaker, 2 – compartment cover of power supply cord, 3 – accumulator compartment cover, 4 – socket for power supply cord connection, 5 – air vents (on both sides), 6 – sealing cup, 7 – control light-emitting diode, 8 – liquid crystal
Control panel is shown in Pic.2. Here: 1 – button «—>», 2 – button «<—», 3 – button «code», 4 – button «input», 5 – control light-emitting diode, 6 – LCD display, 7 – cooling chamber cover. Technical characteristics
Supply voltage DC, V………..12
Difference between ambient temperature and average temperature inside the cooling chamber, °C, not less…………30
Overall dimensions of the container, mm, not more…………. 455х300х410
Volume of working chamber, l, not less……………3,3
Time to steady state, h, not more……………..1
Weight, kg, not more………………20
For better safety of drugs and the prevention of damage of the container is recommended:
To place medical preparations into the container, they should be previosly chilled and packaged.
Do not exploit the container near heat sources and direct sunlight.
Provide free access of air to the air vents when the container is installed.
The refregerator camera must be periodically cleaned with a soft cloth dampened with warm soapy water, and then with clean water followed by wiping it dry.
During the container operation at the bottom of the cooling chamber and the cover may form water droplets (condensate) that must be periodically wiped off.
Do not clean container with powders, pastes, sand, acids and alkalis.Products